Items with full text/Total items : 65275/65275 (100%)
Visitors : 20908286
Online Users : 242
Please use this identifier to cite or link to this item:
|Issue Date: ||2015-07-30 19:01:36 (UTC+8)|
|Abstract: ||短鏈氯化石蠟(short chain chlorinated paraffins, SCCPs) 是指碳鏈長度為10 至13 且氯含量介於30%至72%之間的直鏈氯化烷烴。|
的能力，斯德哥爾摩公約(Stockholm Convention) 目前也擬將短鏈氯
microextraction, VALLME) 為前處理方法，並搭配氣相層析電子捕捉
震盪輔助式液液微萃取法的最佳化條件為：取400 μL 的二氯甲烷
快速注入20 mL 的水樣中，接著使用多功能震盪機以2500 rpm 高速
震盪1 分鐘，再以4500 rpm 的轉速離心3 分鐘，使有機層與水層分
離，隨後取出有機沉澱相並以吹氮回溶至10 μL，最終取1 μL 進樣於
GC/ECNI-MS 進行分析。偵測極限(limit of detection, LOD) 為0.02
μg/L。在inter-days 及intra-day 的測試中，其所得相對標準偏差(RSDs)
樣中所測得的短鏈氯化石蠟濃度為0.08 至0.16 μg/L。
;Short chain chlorinated paraffins (SCCPs) are polychlorinated n-alkanes with carbon chain lengths from 10 to 13 and a chlorination degree between 30 and 72% by mass. SCCPs are resistant to environmental degradation and also have characteristics such as bioaccumulation and long-range transport. A global ban on SCCPs is being considered under the Stockholm Convention on Persistent Organic Pollutants, so it is necessary to develop a simple, efficient and robust method to monitor SCCPs in the environment.
In this study, vortex-assisted liquid-liquid microextraction (VALLME) coupled with GC/ECNI-MS was developed and used to determine SCCPs in water samples. The optimized procedure of VALLME were as follows: 400 μL of dichloromethane was rapidly injected into a 20 mL water
sample. After vortex for 1 min at 2500 rpm and centrifugation for 3 min at 4500 rpm, the mixture became a two-phase solution. Then the sedimented phase was transferred to the vial and evaporated to dryness by a gentle stream of nitrogen. Prior to analysis, 10 μL of acetone containing internal standard was added in the vial and 1μL was injected into
GC/ECNI-MS for analysis.We also used a method termed vortex-assisted enrichment-enhanced liquid-liquid microextraction (VAEELLME) in the analysis of real water samples.With this method, we can obtain higher enrichment and lower detection limit.
The limit of detection (LOD) for SCCPs in water samples was determined to be 0.02 μg/L. The relative standard deviations (RSDs) of inter-days and intraday tests were lower than 9% and the SCCPs levels in water samples ranged from 0.08 to 0.16 μg/L.
|Appears in Collections:||[化學研究所] 博碩士論文|
Files in This Item:
All items in NCUIR are protected by copyright, with all rights reserved.
::: Copyright National Central University. | 國立中央大學圖書館版權所有 | 收藏本站 | 設為首頁 | 最佳瀏覽畫面: 1024*768 | 建站日期：8-24-2009 :::